Sigh: one deep breath per 80 controlled respirations, the inspiration time is 1.5 times of the setting point and the permissible error is -10% or 15%.
Pressure limit range:1.0~ 6.0kPa
Airway pressure alarm: Audible and visual and with yellow and red color indicatingupper alarm: 0.9 kPa ~ 5.4 kPa
Ventilation Volume alarm: ≥ 18L/ MIN
Power Supply Alarm"Ac/dc power supply are after failing to send out alarm immediately
Alarm Time: keep >120s"
The airway pressure continues to be higher than 15 hPa ±1 hPa for 15s±1s, then the machine will raise an audible alarm, the pressure will be displayed in red and the continuous high pressure red alarm wording is displayed on the screen of the anesthetic respirator.
Ambient temperature: 10 ~ 40oC
Relative humidity:no higher than 80 %
Atmospheric pressure:860 hPa ~ 1060 hPa
Power requirement: 100-120 Vac, 50/60 Hz;
Air source requirement: Medical oxygen and laughing gas with a rated pressure ranging from 0.3 to 0.5MPa.
Attention: the AC power supply used for the anesthesia machine must be well grounded.
Attention: the anesthesia machine used must be equipped with a carbon dioxide monitor complying with ISO 9918:1993, an oxygen monitor complying with ISO 7767:1997 and an expiratory gas volume monitor complying with 188.8.131.52 of Medical Electrical Equipment Part II: Special Requirements for Safety and Basic Performance of Anesthesia System.
Ambient temperature: -15oC ~ +50oC
Relative humidity: no higher than 95 %
Atmospheric pressure: 86 kPa ~ 106 kPa.
It should be stored in a room without corrosive gas and well ventilated
packaging boxcomply with the requirement of GB/T 15464
Between the packaging box and product, soft material with appropriate thickness provided to prevent loosening and mutual friction during transportation
Moisture protection and rain protection to ensure that product is protected from natural damage.
Safty & Alarm
Gas source pressureImmediate alarm;“high level alarm” indicator flickers raising an audible alarm.
Airway Pressure AlarmThe airway pressure exceeds 15hpa. The maximum delay is no longer than 17s. The “high level alarm” indicator flickers raising an audible alarm.
Tidal Volume AlarmLower than 50ml, The alarm will be raised after one breathing cycle.
Ventilation Volume AlarmLower: 0~12L/Min; Up: 18L/ Min
Battery Voltage AlarmIt alrms when the battery voltage is lower than 10.5V.
Power On Self TestSelf test before working, assure it's working normally.
Top Quality: We have strict quality control system and enjoy good reputation in the market.
Excellent Service: We treat clients as friend and aim at building long term business relatiionship.
OEM Accepted : We Can print your own logo on the products.
Unbeatable Price: We always ensure high price performance to benefit our customers.
Timely Delivery:Your order shall be shipped once finished .
After-Sale Service:All Products Have One Year Warranty.